Valves, Transistors, IC's and Components. Bench test techniques, Test Gear - Valve Testers, Scopes, CRT Testers, Bridges, Sig Gens, Pattern Gens, Workshop tours etc
- Posts: 188
- Joined: Sat Nov 22, 2014 9:43 pm
Hi from RR. Does anyone know if it's ok to use a 240 volt variac on 110 volt? Or wouldn't it work? I'm currently trying to repair American jukebox amplifiers which are all DC coupled transistor type and as volatile as can be and some of the transistors are only available from the states and quite pricey and I've already blown some, not new ones thank goodness. And I've got a 240 volt 2 amp variac which I would like to use to power up the amp's slowly to start with after a rebuild, I'm currently using a yellow CTE type tool transformer to provide 110 volt supply for the amp's which normally run from a dirty great big step down transformer in the jukebox. And does anyone here know what is the smallest variac you can get? The smallest I've ever seen is 0.8 amp. RR.
- Posts: 151
- Joined: Sat Feb 11, 2012 8:35 pm
A variac is simply an adjustable transformer. It allows you to set the output anywhere from 0v to the supply voltage (approx. 240v in UK). Some variacs have extra turns. Depending on how it's wired up, you can get slightly more than 240v (typically 270v) if the extra turns are used.
If you supply your variac with 110v, it will adjust over the range of 0-110v. There's no problem about using a 240v variac on a lower voltage. But be careful not to draw more than the maximum current allowed from your variac. If your variac is rated 240v 2 amps and you supply it with 110v, you can still only draw a maximum 2 amps from it at 110v, which means it can only supply about half as much power.
If you just want to slowly power up an American amplifier, you can connect your variac to 240v, set it to zero and slowly turn the dial until you reach about halfway or 110v on the scale. Just be careful not to go above halfway. No additional transformer is needed unless you need an isolating transformer. In that case, you can connect your yellow building site transformer to the output of the variac. The output from the transformer will vary from 0v at minimum to 110v when the variac is set to maximum.
Return to In The Workshop
Who is online
Users browsing this forum: Bing [Bot] and 2 guests